Caterina Carboni, Lorenzo Bisoni, Nicola Carta and Massimo Barbaro
View Full Paper
 T. Boretius, A. Pascual-Font, E. Udina, T. Stieglitz,and X. Navarro, “Biocompatibility of chronically im-planted transverse intrafascicular multichannel elec-trode (time) in the rat sciatic nerve,” IEEE Transac-tion on Biomedical Engineering, vol. 58, no. 8, pp.2324–2332, 2011.
 F. Shahrokhi, K. Abdelhalim, D. Serletis, P. L. Carlen,and R. Genov, “The 128-channel fully differentialdigital integrated neural recording and stimulationinterface,” IEEE Trans. Biomedical Circuit System,vol. 4, pp. 149–161, 2010.
 J. Lee, H. G. Rhew, D. R. Kipke, and M. P. Flynn, “A64 channel programmable closed-loop neurostimula-tor with 8 channel neural ampliﬁer and logarithmicadc,” IEEE Journal of Solid-State Circuits, vol. 45,pp. 1935–1945, 2010.
 D. Loi, C. Carboni, G. Angius, G. Angotzi, M. Bar-baro, L. Raffo, S. Raspopovic, and X. Navarro, “Pe-ripheral neural activity recording and stimulation sys-tem,” Biomedical Circuits and Systems, IEEE Trans-actions on, vol. 5, no. 4, pp. 368–379, 2011.
 W. Wattanapanitch and R. Sarpeshkar, “A low-power32-channel digitally programmable neural recordingintegrated circuit,” IEEE Transaction on BiomedicalEngineering, vol. 5, no. 6, pp. 593–602, 2011.
 R. R. Harrison, R. J. Kier, C. A. Chestek, V. Gilja,P. Nuyujukian, S. Ryu, B. Greger, F. Solzbacher, andK. V. Shenoy, “Wireless neural recording with singlelow-power integrated circuit,” IEEE Transactions onNeural Systems and Rehabilitation Engineering, pp.322–329, 2009.
 R. Rieger, J. Taylor, A. Demosthenous, N. Donaldson,and P. J. Langlois, “Design of a low-noise preampli-ﬁer for nerve cuff electrode recording,” IEEE Journalof Solid-State Circuits, vol. 38, no. 8, pp. 1373–1379,2003.
 H. Gao, R. Walker, P. Nuyujukian, K. Makinwa,K. Shenoy, B. Murmann, and T. Meng, “Hermese:A 96-channel full data rate direct neural interface in0.13 µm cmos,” IEEE Journal of Solid-State Circuits,vol. 47, no. 4, pp. 1043–1055, 2012.
 S. Micera, J. Carpaneto, and S. Raspopovic, “Con-trol of hand prostheses using peripheral information,”vol. 3, 2010, pp. 48–68.
 A. Zabihian and A. Sodagar, “A new architecture formulti-channel neural recording microsystems basedon delta-sigma modulation,” IEEE Conference onBiomedical Circuits and Systems, pp. 81–84, 2009.
 S. Y. Lee and S. C. Lee, “An implantable wire-less bidirectional communication microstimulator forneuromuscolar stimulation,” IEEE Trans. Circuit Sys-tem, vol. 52, pp. 2526–2538, 2005.
 W. Grill and J. Mortimer, “Stimulus waveformsfor selective neural stimulation,” Engineering inMedicine and Biology Magazine, IEEE, vol. 14, no. 4,pp. 375 –385, jul/aug 1995.
 T. Tanzawa and T. Tanaka, “A dynamic analysis of thedickson charge pump circuit,” Solid-State Circuits,IEEE Journal of, vol. 32, no. 8, pp. 1231–1240, Aug.
 T. Yamazoe, H. Ishida, and Y. Nihongi, “A chargepump that generates positive and negative high volt-ages with low power-supply voltage and low powerconsumption for non-volatile memories,” in Circuitsand Systems, 2009. ISCAS 2009. IEEE InternationalSymposium on, May, pp. 988–991.
 C. Carboni, D. Loi, and M. Barbaro, “A front-endstage for neural signal recording based on a sigmadelta modulator,” Proceedings of the internationalconference on Biomedical Electronics and Devices(BIODEVICES2012), pp. 207–212, February 2012.
 R. Schreier and T. Gabor C., “Understandingdelta sigma data converters,” IEEE Press/Wiley-Interscience, 2001.70
 C. Carboni, N. Carta, M. Barbaro, and L. Raffo, “Asigma-delta architecture for recording of peripheralneural signals in prosthetic applications,” Biomedi-cal Robotics and Biomechatronics (BioRob), 2012 4thIEEE RAS EMBS International Conference on, pp.448 –453, June 2012.
 X. Liu, A. Demosthenous, and N. Donaldson, “An in-tegrated implantable stimulator that is fail-safe with-out off-chip blocking-capacitors,” Biomedical Cir-cuits and Systems, IEEE Transactions on, vol. 2,no. 3, pp. 231 –244, sept. 2008.